Hi, I'm using a U3 with latest firmware in a project, developing with NI LabWindows / CVI. Everything is working well.
My only issue is with error handling. I'm using the string function ErrorToString to decode the error numbers. It works fine. However, I have noticed that when I "pull the plug" of the LabJack during use, the errors I always get are a "invalid handle" rather than what I would expect: "labjack not found".
Regardless, to attempt to recover from these kinds of big errors, I attempt a ResetLabJack() but that doesn't seem to work either.
Any thoughts on this?