EI1050 Labview example
2 replies to this topic
Posted 13 November 2012 - 05:07 AM
I have an EI1050 probe and it works fine with your program Labjack Python EI1050 example, but when I run labview example "U3 Single EI-1050 Example" I get this error "LabJack Error #52: CRC failed for SHT communication occured at LJUD_GetNextResult.vi" Why is this?
Posted 13 November 2012 - 08:10 AM
Strange. Try running with the lightbulb on, to slow down LabVIEW, and see if that makes a difference. I do see that the LabVIEW example uses FIO6 for Red/Brown, while the python utility uses FIO7, so make sure you move those wires. You could also try changing the LV example to use FIO7 the same as the python utility.
Posted 14 November 2012 - 01:44 AM
Thank you very much for your quick answer and sorry, I did not realize the change in FIO6 and FIO7. Now both of them work-
0 user(s) are reading this topic
0 members, 0 guests, 0 anonymous users